- Open Access
Cell surface and cell outline imaging in plant tissues using the backscattered electron detector in a variable pressure scanning electron microscope
© Talbot and White; licensee BioMed Central Ltd. 2013
- Received: 12 July 2013
- Accepted: 4 October 2013
- Published: 17 October 2013
Scanning electron microscopy (SEM) has been used for high-resolution imaging of plant cell surfaces for many decades. Most SEM imaging employs the secondary electron detector under high vacuum to provide pseudo-3D images of plant organs and especially of surface structures such as trichomes and stomatal guard cells; these samples generally have to be metal-coated to avoid charging artefacts. Variable pressure-SEM allows examination of uncoated tissues, and provides a flexible range of options for imaging, either with a secondary electron detector or backscattered electron detector. In one application, we used the backscattered electron detector under low vacuum conditions to collect images of uncoated barley leaf tissue followed by simple quantification of cell areas.
Here, we outline methods for backscattered electron imaging of a variety of plant tissues with particular focus on collecting images for quantification of cell size and shape. We demonstrate the advantages of this technique over other methods to obtain high contrast cell outlines, and define a set of parameters for imaging Arabidopsis thaliana leaf epidermal cells together with a simple image analysis protocol. We also show how to vary parameters such as accelerating voltage and chamber pressure to optimise imaging in a range of other plant tissues.
Backscattered electron imaging of uncoated plant tissue allows acquisition of images showing details of plant morphology together with images of high contrast cell outlines suitable for semi-automated image analysis. The method is easily adaptable to many types of tissue and suitable for any laboratory with standard SEM preparation equipment and a variable-pressure-SEM or tabletop SEM.
- Backscattered-electron imaging
- Plant cell outlines
- Image analysis
- Arabidopsis thaliana
The analysis of developmental changes in plant cells, tissues and organs often requires quantification of subtle alterations in cell morphology. Measurements of cell size and shape require contrast enhancement of cell boundaries (cell walls or plasma membrane) to produce high-contrast images suitable for subsequent analysis. Methods to enhance plant cell outlines vary in complexity from straightforward imaging of cell wall autofluorescence to lengthy, multistep processing for three-dimensional analysis of tissue architecture by confocal laser scanning microscopy (CLSM) e.g., [1–3]. These methods are often suited only to particular plants, tissues or cell types due to inherent differences in cell or tissue properties across different species.
Some simple methods, such as differential interference contrast of cleared tissue e.g., , produce relatively low-contrast images unsuitable for automated image analysis. Cell outline contrast can be increased by staining fresh tissue with, for example, propidium iodide e.g., [5–7], or membrane-binding FM dyes e.g., [8, 9] but these stains do not easily penetrate all tissues without pre-treatment, particularly aerial parts of the plant, which are often coated with a waxy cuticle. CLSM can also be used to detect green fluorescent protein (GFP) targeted to the cell surface e.g., , but it is not possible to obtain GFP transformants for every plant or tissue under study. In addition, there is often a requirement with CLSM to acquire a 3-dimensional image series to obtain a complete view of tissues with complex shapes. Considerable subsequent computation is then required to extract information about a single layer of cells such as the epidermis from these stacks [5, 11, 12].
The Scanning Electron Microscope (SEM) has seldom been used to generate images for the purposes of analysis, largely because conventional imaging of biological tissue under high vacuum SEM requires coating the tissue with a conductive metal, which obscures information in the sample irrespective of the tissue and different beam energies . The images obtained provide useful information about overall tissue morphology and surface details, but most analysis packages struggle to correctly discriminate cell outlines using the subtle differences in grey levels in these pseudo-3D images. SEM imaging usually involves detection of secondary electrons (SE), which are sample-derived electrons generated from interaction of the primary electron beam with the top 1–10 nm of the sample surface [13, 14]. In contrast, backscattered electrons (BSEs) are beam electrons which have been scattered deeper within the sample. BSEs can provide atomic number contrast in which differences in signal intensity are related to local differences in the average atomic number .
In an environmental pressure SEM (EP-SEM) or variable-pressure SEM (VP-SEM), the specimen chamber operates at much lower vacuum due to the presence of an 'imaging gas’ (typically nitrogen). The gaseous environment around the sample helps to reduce charging artefacts at the sample surface , and the specimen can be viewed uncoated or in the case of EP-SEM where water is the imaging gas, viewed hydrated with no processing. In a VP-SEM, SE and BSE signals provide a flexible range of options to image biological tissues [15, 16] and can reveal detail not previously visible in coated tissue under high vacuum.
Previously we used a BSE detector with VP-SEM to produce images of high contrast cell outlines in uncoated, critical point dried barley leaves for image analysis . In this paper we extend this technique to a wider range of plant tissues, describe how to optimise this protocol and apply it to quantify cell size in leaves of the model plant Arabidopsis thaliana. The advantages of this protocol are that it is simple and quick, it enables recording of surface details together with high contrast images for quantitative analysis using freely available software, and is suitable for any laboratory with standard SEM preparation equipment and any VP-SEM, including tabletop models.
Optimising cell wall outlines in A. thaliana leaves with the BSE detector
To extend BSE imaging further, we optimised parameters for producing high contrast images of cell outlines suitable for analysis of cell size and shape. We focused on A. thaliana, a model species for studying dicotyledon plant growth and development, but we also included common dicotyledon (cotton) and monocotyledon (barley, wheat, rice and Brachypodium) species for comparison. Leaves are ideal for this type of analysis as they are relatively flat, and epidermal cells generally contain little cytoplasm and few chloroplasts, components which add to the BSE signal and complicate image analysis.
Spot size, working distance, and chamber pressure
Low temperature VP-SEM and extended-pressure-SEM (EP-SEM)
BSE imaging of other plant tissues
Origin of bright cell wall outlines
In A. thaliana leaves, the strongest peaks detected were from calcium, phosphorus and sulphur with smaller contributions from magnesium and potassium (Figure 9A), all of which are normally present in plant tissues [21, 22]. Since there is generally a significant proportion of calcium in cell walls, we tested its contribution to the BSE signal by chelation of bound calcium with EDTA; this treatment resulted in a loss of BSE signal from epidermal cell walls (Figure 9C cf. Figure 9B), and loss of calcium peaks from the EDS spectra (Figure 9A). Note that EDTA treatment also resulted in loss of the magnesium peak (Figure 9A), since EDTA chelates both cations . In comparison, EDS analysis of barley leaves suggested that there is a strong contribution from potassium in this tissue (Additional file 2). Collection of x-ray maps (to correlate BSE signal with element distributions) was not informative, since resolution is low and very long acquisition times (> 2 h) are required, which results in specimen damage.
The magnification selected to capture images for analysis depends on how many cells can be accurately outlined and measured by the software. In this case images of A. thaliana leaves were taken at 200× magnification at a resolution of 1024 × 768 pixels, from which 30–40 cells were measured. Larger areas may be analysed if images from adjacent areas are stitched together beforehand, or the image is captured at higher resolution; we recommend testing several magnifications at different image resolutions to determine optimal image capture settings.
It must be noted that stomatal guard cells were lost from these images during processing. If stomatal cells are to be included in the analysis, a more detailed processing procedure should be developed, since guard cell walls are much less bright than surrounding pavement epidermal cell boundaries (Additional file 6). For analysis it is also important to avoid wrinkles in the walls (as a result of tissue shrinkage) as these will add to the signal and produce artefactual cell 'boundaries’ when processing the image for analysis. Critical point drying directly after methanol fixation and transfer to ethanol minimises such artefacts .
Comparison with other methods to highlight cell outlines
A. thaliana plants expressing GFP targeted to the cell surface (Figure 11C and D) and staining with cell wall binding dyes, such as propidium iodide (Figure 11E) can also be used to detect cell outlines. However, A. thaliana cotyledons and leaves are rarely flat and a Z-series must be collected to overcome this topography, and it is then difficult to distinguish between cell types in a maximum projection image derived from a Z stack showing cell surface GFP (Figure 11C). The cell walls of young A. thaliana cotyledons appeared to stain readily with propidium iodide (Figure 11E), but true leaves stained only after initial vacuum treatment (Additional file 7; ). Since the dye did not easily penetrate below the epidermis, this has the advantage of avoiding confusion between cell layers. Nevertheless, although contrast was high in individual sections of a Z-stack (Figure 11E; Additional file 7A), fluorescence from the periclinal wall of the epidermis reduced contrast of the anticlinal walls in maximum Z-stack projections (e.g., Additional file 7B).
Cell wall autofluorescence can also be used to obtain cell outlines. While A. thaliana leaf tissue exhibits little cell wall autofluorescence, cereal leaves contain brightly autofluorescent wall components . Outlines are readily detected under UV excitation, but as above, a Z-series or image stitching may be required to overcome tissue topography. Since the epidermis is more autofluorescent than the underlying cells however, a relatively clear image of epidermal cell outlines was obtained (Figure 11F).
Variable pressure (VP)-SEMs allow detection of signals under low vacuum, enabling the use of minimally-processed, uncoated tissues. This study showed that imaging uncoated samples allowed detection of cell outlines with the BSE detector, information which is difficult to obtain using VP-SE or conventional high vacuum SE detectors. The technique outlined here also overcomes problems encountered when attempting to resolve cell outlines using other imaging methods.
Cell wall outline imaging with the BSE detector
Fix tissue in methanol for 10 min or longer. Vacuum infiltrate if necessary.
Dehydrate further in dry ethanol for 1 h (small tissues) or overnight (large tissues).
Critical point dry following manufacturer’s recommendations.
Mount tissue on SEM stub and observe as soon as possible (same or next day).
Recommended microscope operating conditions:
Accelerating voltage = 20 kV (10 kV for surface topography)
Chamber pressure = 10–40 Pa
Working distance = 7 mm (check optimum distance for your detector)
Spot size = 0.7 nA
Increase image contrast to enhance cell wall outline contrast
If charging is a problem, ensure good contact of tissue with the stub and apply carbon paste to the edges of the tissue. If charging remains, coat tissue with carbon.
Store tissue in a desiccator or low humidity cabinet.
Origin of cell wall outline contrast with the BSE detector
Backscattered electrons contribute to both of the contrast mechanisms underlying image formation in the SEM; compositional (atomic number) contrast and topographic contrast . Compositional contrast most likely explains the bright signal from cell walls in most tissues observed in this study. Plant cell walls contain varying amounts of calcium, phosphorous, silicon, sulphur, potassium, magnesium and chloride, depending on the species and tissue [21, 22]. These components produce higher BSE yields compared to lower atomic number organic constituents (carbon, hydrogen and oxygen) in the cell wall and cytoplasm . Some cell walls normally accumulate ions, for example, trichome walls (e.g., A. thaliana; ), which increases endogenous BSE contrast (e.g., Figure 1B). X-ray microanalysis data suggested that calcium in A. thaliana (Figure 9) and potassium in barley (Additional file 2) leaf tissue were the main constituents underlying the strong BSE signals in epidermal cell walls.
Calcium is a likely candidate as a source of BSE signal at cell junctions. Plant cell walls preferentially accumulate cations, since carboxyl groups on demethylated pectin, and to a lesser extent on cellulose and proteins, impart an overall negative charge [30, 31]. Calcium is normally bound to demethylated pectin in walls, and is enriched at cell junctions, which strengthens cell-cell adhesion . Interestingly, it has recently been shown that propidium iodide competes with calcium in binding to carboxyl groups on demethylated pectin , explaining why cell wall outlines revealed by propidium iodide (Figure 11E and Additional file 7) are very similar to those observed with the BSE detector (e.g., Figure 2C). Specimen processing results in leaching and relocation of un-bound ions from cells  and it is likely that cations not originally located in the wall, including calcium, magnesium and potassium, accumulate at unoccupied anionic sites within the wall during preparation of tissues for SEM. In this way, removal of water may create additional compositional contrast at wall boundaries for BSE imaging.
Although plant tissues are optimally preserved for SEM in the frozen state , and EP-SEM is beneficial for imaging certain tissues , no bright wall outlines could be seen in frozen tissues or in fresh tissues observed with EP-SEM (Figure 6). Epidermal cell outlines were visible, but these were generated by topography of epidermal cells, and were of low contrast compared to outlines observed in critical point-dried tissue (e.g., Figure 2C). Furthermore, there are disadvantages to using either low temperature VP-SEM or EP-SEM imaging to obtain images of cell outlines. A common drawback is that imaging with either method needs to be quick, as the tissue will freeze-dry due to sublimation during imaging of frozen tissue (without a dedicated liquid nitrogen-cooled cryo-SEM stage). Tissue also loses water rapidly when imaging with EP-SEM, and tissues viewed by either method are sensitive to beam damage. A final disadvantage is that the tissue cannot be stored and re-imaged if required. Nevertheless, imaging frozen tissue either with  or without  a Peltier-cooled stage may be of use to quickly examine uncoated tissues in the VP-SEM without the need for the dedicated and expensive cryo-preparation equipment required for longer analysis of tissue held at close to liquid nitrogen temperatures.
Comparison with other techniques for visualizing cell outlines
CLSM images yield high-contrast cell outlines only when sufficient fluorescence can be obtained from cell walls or membranes, either from staining or localised GFP expression. The waxy cuticles found on most plant epidermal surfaces are generally quite hydrophobic, and therefore commonly used aqueous stains, including propidium iodide, may not penetrate to stain periclinal cell walls without vacuum infiltration or pre-treatment to remove some surface wax. However, such treatments must not be so harsh that cell membrane integrity is compromised, since for high contrast, cell wall stains must be retained in the apoplastic space, and excluded from the cytoplasm.
Topography is often a problem because confocal images are generated from a thin optical slice of tissue, and information from a complete 3D surface can generally be obtained only from a z-stack. However, isolating epidermal fluorescence from such stacks is time-consuming. Topography may be overcome by stitching adjacent images from slightly different focal planes, but this is also time-consuming and may require manual checking even if image capture (with autofocus) and draft stitching can be automated.
Many plant tissues show cell wall autofluorescence, and high contrast cell outlines can be obtained from many cereal tissues, which generally show strong blue-green fluorescence with UV excitation . However, even if present, cell wall autofluorescence may be insufficient for the high contrast outlines required for image analysis, and BSE images provide superior contrast.
One potential problem with SEM observation, particularly in relation to image quantification, is that processing tissue through fixation, dehydration and critical point drying (CPD) can lead to tissue shrinkage [18, 19], and changes in cell size. We have found that concomitant fixation and dehydration in 100% methanol followed by transfer to ethanol prior to CPD resulted in the least tissue shrinkage and best preservation of tissue morphology . An advantage of this fixation method is that it is very quick; many tissues can be processed for imaging within 2–3 h. Original tissue dimensions will be preserved as faithfully as possible if the tissue is viewed soon after processing, and stored in a desiccated or low-humidity environment for future imaging if necessary.
As noted earlier, most epidermal surfaces have surface elaborations such as waxy cuticles or mineral deposits. In some cases, the primary electron beam can penetrate these coverings (e.g. Figure 7D), but heavily elaborated tissues cannot be analysed this way (Figure 7F,H). Furthermore, additional SE or BSE signal from organelles and other cytoplasmic structures (Figure 7B,D; ) may interfere with the ability to capture clear cell wall outlines. Another limitation is that only relatively flat tissue is suitable for imaging in order that cell size is faithfully represented. However, it is possible to orient tissue on the SEM stub or rotate the stage in order to image the epidermal area of interest. As with all preparation techniques it is advisable to first assess the suitability of this method for the cells or tissues of interest.
The technique presented here for obtaining high contrast images suitable for analysis of epidermal cell size and shape is relatively quick and simple, and with the rising popularity of affordable desktop SEMs, this protocol provides a good alternative to other imaging methods. Fortuitously, BSE imaging of cell outlines is well-suited to epidermal cells since they generally contain large vacuoles and have little cytoplasm with few organelles. There are very many image analysis packages and protocols available for processing images; the processing and analysis steps shown here for ImageJ/Fiji can be readily adapted to an institution’s preferred analysis package.
For many plant tissues, quantification of cell surface size and shape can be done rapidly using the protocol outlined above with relatively few artefacts. Imaging uncoated tissue in the variable-pressure SEM using the BSE detector is straightforward and provides a simple protocol for laboratories with standard SEM processing equipment. Furthermore, tissues can be processed in batch, examined and stored for future imaging if required.
Tissues from a number of different plants were prepared for SEM, including Arabidopsis thaliana (L.) Heynh (Columbia), Brachypodium distachyon L. (21–3 line), Gossypium barbadense L. (cotton; Pima variety), Gossypium hirsutum L. (cotton; Coker variety), Hordeum vulgare L. (barley; Himalaya cultivar), Oryza sativa L. (rice; Nipponbare), and Triticum aestivum L. (wheat; Bobwhite). Tissues were dissected from the plant and immediately fixed in 70% ethanol for a minimum of 30 min, or 100% anhydrous methanol for 10 min [20, 37], at room temperature. Application of light vacuum within the first 5 min (or until tissue sank) improved ethanol fixation of more difficult tissues (e.g., rice leaves), although tissue sank much faster in methanol even without vacuum treatment. For ethanol fixation, tissues were dehydrated to 100% anhydrous ethanol in 10% steps, 30 min each step (100% ethanol was changed twice, 30 min each). For methanol fixation, methanol was replaced with 100% anhydrous ethanol twice, 30 min each; larger tissue pieces were left overnight in ethanol . After a rinse in 100% anhydrous ethanol, tissues were critical point dried with an Autosamdri-815 automatic critical point drier (Tousimis Research Corporation, Rockville USA). Artefacts, such as shrinkage of tissue and cell wall wrinkling, were minimized by methanol fixation [20, 37] and by processing tissue straight through to critical point drying within 1–2 hours of reaching the second 100% ethanol change, or the day after fixation and dehydration at the latest. Methanol fixation is recommended as we generally found it to be superior to ethanol and other commonly used SEM preparation procedures .
Specimens were mounted on aluminium stubs with double-sided sticky carbon tabs (ProSciTech, Qld, Australia) and visualized uncoated in a Zeiss EVO LS 15 Extended Pressure-Scanning Electron Microscope (Carl Zeiss Pty Ltd, Sydney, Australia) in variable-pressure (VP) mode (with nitrogen as the imaging gas), with a final VP aperture of 100 μm. The backscattered electron detector was a 4-quadrant solid state type mounted below the final aperture directly above the sample. Other instrument settings are detailed in the text and figure captions. If charging was excessive between 10 and 50 Pa, tissue was removed from the chamber and carbon paste was applied to the edges, improving contact between sample and stub. If charging was still present at up to 100 Pa the tissue was coated with carbon (~30-40 nm) using an Emitech K500X sputter coater with K250 carbon coating attachment (Quorum Emitech, Kent, UK). For comparison, some tissue pieces were sputter coated with gold (~20 nm). To reduce absorption of moisture or further changes in tissue dimensions, critical point dried tissue was stored (mounted or un-mounted) in a electronic humidity-controlled storage cabinet set at 10% relative humidity (Thermoline Scientific, Australia).
Low temperature VP-SEM and extended-pressure-SEM (EP-SEM)
For low-temperature VP-SEM, leaves were dissected from the plant and immediately mounted on a drop of water on a 9 mm stub with a double-sided sticky carbon tab. The stub and tissue were frozen in liquid nitrogen then transferred to the Deben Coolstage which had been pre-cooled to -20-30°C . This procedure provides approximately 20 min imaging time before the tissue freeze-dries in the vacuum due to sublimation of water. A dedicated cryo-stage which enables imaging of tissue at liquid nitrogen temperatures is best, but such dedicated equipment is not easily accessible or affordable for most laboratories.
For extended pressure SEM (EP-SEM), leaves from 14-day old agar-grown A. thaliana Columbia seedlings were dissected and immediately mounted on a 9 mm stub with a double-sided sticky carbon tab. Several small drops of distilled water were placed around the tissue to maintain local humidity, and the stub was transferred to a Deben Coolstage (Peltier-cooled stage; Deben, UK) attached to the Zeiss EVO LS15 and the chamber pumped down. To maintain tissue in a hydrated state, EP-SEM conditions were 82% humidity, 600–700 Pa chamber pressure, 2-3°C Peltier stage temperature, and 20–25 kV accelerating voltage.
For X-ray microanalysis, critical point dried 14-day old agar-grown A. thaliana Columbia leaves and barley leaf pieces were analyzed with an Oxford Inca PentaFetx3 SiLi detector with a 30 mm2 ATW2 window (Oxford Instruments). Leaves were carbon coated to avoid excessive charging (see above) and analyzed under high vacuum using 20 kV accelerating voltage (1.7 nA probe current; 150 μA beam current) at 18-20 mm working distance and approximately 400× magnification. Spectra were acquired over 2 min; peaks were manually confirmed in the software (INCA suite v. 4.11). Background spectra from different areas of the stub (carbon tab) were acquired under the same conditions for comparison. To test the contribution of calcium to the BSE signal, freshly harvested A. thaliana leaves were extracted with 1% ethyelenediaminetetracetic acid (EDTA, sodium salt). Leaves were first vacuum infiltrated with the solution and left overnight. Pieces were washed in water, fixed in methanol and dehydrated in ethanol (see above), and critical point dried. Control leaves were fixed in methanol, dehydrated in ethanol and critical point dried.
Light and Confocal microscopy
For Differential Interference Contrast (DIC) imaging, leaves from 14-day old agar-grown A. thaliana Columbia seedlings were cleared in saturated chloral hydrate overnight. Cleared leaves were mounted in 50% glycerol and observed with a Zeiss Axioimager M1 compound microscope using DIC optics. For fluorescence visualization of cell outlines in A. thaliana, cotyledons from 7-day old agar-grown Columbia seedlings were dissected from the plant and mounted directly in 10 μg/ml propidium iodide. After 10 min they were observed on a Leica TCS SP2 CLSM using 488 nm excitation and 560–620 nm emission. Similarly, A. thaliana leaves from 3 week old seedlings were cut at the petiole and infiltrated with 100 μg/ml propidium iodide under light vacuum , for 3 × 1 min. For visualization of GFP in A. thaliana line 29–1 (plasma-membrane localized GFP; ), leaves were dissected from 3–4 week old agar-grown or soil-grown seedlings, mounted in water and observed using 488 nm excitation and 500–530 nm emission. Barley leaf pieces were dissected and mounted in silicone oil, and autofluorescence between 420–580 nm was detected following UV (405 nm) excitation.
- Bougourd S, Marrison J, Haseloff J: An aniline blue staining procedure for confocal microscopy and 3D imaging of normal and perturbed cellular phenotypes in mature Arabidopsis embryos. Plant J. 2000, 24 (4): 543-550. 10.1046/j.1365-313x.2000.00892.x.View ArticlePubMedGoogle Scholar
- Truernit E, Bauby H, Dubreucq B, Grandjean O, Runions J, Barthelemy J, Palauqui J-C: High-resolution whole-mount imaging of three-dimensional tissue organization and gene expression enables the study of phloem development and structure in Arabidopsis. Plant Cell. 2008, 20: 1494-1503. 10.1105/tpc.107.056069.PubMed CentralView ArticlePubMedGoogle Scholar
- Wuyts N, Palaqui J-C, Conejero G, Verdeil J-L, Granier C, Massonnet C: High-resolution three-dimensional imaging of the Arabidopsis leaf enables the analysis of cell dimensions in the epidermis and mesophyll. Plant Methods. 2010, 6 (July): 1-14.Google Scholar
- Perez-Perez JM, Rubio-Diaz S, Dhondt S, Hernandez-Romero D, Sanchez-Soriano J, Beemster GTS, Ponce MR, Micol JL: Whole organ, venation and epidermal cell morphological variations are correlated in the leaves of Arabidopsis mutants. Plant Cell Environ. 2011, 34: 2200-2211. 10.1111/j.1365-3040.2011.02415.x.View ArticlePubMedGoogle Scholar
- Fitzgibbon J, Beck M, Zhou J, Faulkner C, Robatzek S, Oparka K: A developmental framework for complex plasmodesmata formation revealed by large-scale imaging of the Arabidopsis leaf epidermis. Plant Cell. 2013, 25 (1): 57-70. 10.1105/tpc.112.105890.PubMed CentralView ArticlePubMedGoogle Scholar
- Kanaoka MM, Pillitteri LJ, Fuji H, Yoshida Y, Bogenschutz NL, Takabayashi J, Zhu J-K, Torii KL: SCREAM/ICE1 and SCREAM2 specify three cell-state transitional steps leading to Arabidopsis stomatal differentiation. Plant Cell. 2008, 20: 1775-1785. 10.1105/tpc.108.060848.PubMed CentralView ArticlePubMedGoogle Scholar
- Rounds CM, Lubeck E, Helper PK, Winship LJ: Propidium iodide competes with Ca2+ to label pectin in pollen tubes and Arabidopsis root hairs. Plant Physiol. 2011, 157: 175-187. 10.1104/pp.111.182196.PubMed CentralView ArticlePubMedGoogle Scholar
- Fernandez R, Das P, Mirabet V, Moscardi E, Traas J, Verdeil J-L, Malandain G, Godin C: Imaging plant growth in 4D: robust tissue reconstruction and lineaging at cell resolution. Nat Methods. 2010, 7 (7): 547-553. 10.1038/nmeth.1472.View ArticlePubMedGoogle Scholar
- Zhang C, Mallery EL, Schlueter J, Huang S, Fan Y, Brankle S, Staiger CJ, Szymanski DB: Arabidopsis SCARs function interchangeably to meet actin-related protein 2/3 activation thresholds during morphogenesis. Plant Cell. 2008, 20: 995-1011. 10.1105/tpc.107.055350.PubMed CentralView ArticlePubMedGoogle Scholar
- Cutler SR, Erhardt DW, Griffitts JS, Somerville CR: Random GFP: cDNA fusions enable visualization of subcellular structures in cells of Arabidopsis at a high frequency. PNAS. 2000, 97 (7): 3718-3723. 10.1073/pnas.97.7.3718.PubMed CentralView ArticlePubMedGoogle Scholar
- Reddy GV, Heisler MG, Ehrhardt DW, Meyerowitz EM: Real-time lineage analysis reveals oriented cell divisions associated with morphogenesis at the shoot apex of Arabidopsis thaliana. Development. 2004, 131: 4225-4237. 10.1242/dev.01261.View ArticlePubMedGoogle Scholar
- de Reuille PB, Bohn-Courseau I, Godin C, Traas J: A protocol to analyse cellular dynamics during plant development. Plant J. 2005, 44: 1045-1053. 10.1111/j.1365-313X.2005.02576.x.View ArticlePubMedGoogle Scholar
- Stokes DJ: Principles and practice of variable pressure/environmental scanning electron microscopy (VP-SEM). 2008, UK: John Wiley and Sons Ltd.View ArticleGoogle Scholar
- Goldstein J, Newbury D, Joy D, Lyman C, Echlin P, Lifshin E, Sawyer L, Michael J: Scanning electron microscopy and X-Ray microanalysis 3rd Ed. 2003, USA: SpringerView ArticleGoogle Scholar
- Collins SP, Pope RK, Scheetz RW, Ray RI, Wagner PA, Little BJ: Advantages of environmental scanning electron microscopy in studies of microorganisms. Microsc Res Tech. 1993, 25: 398-405. 10.1002/jemt.1070250508.View ArticlePubMedGoogle Scholar
- Stabentheiner E, Zankel A, Polt P: Environmental scanning electron microscopy (ESEM) – a versatile tool in studying plants. Protoplasma. 2010, 246: 89-99. 10.1007/s00709-010-0155-3.View ArticlePubMedGoogle Scholar
- Greenup AG, Sasani S, Oliver SN, Talbot MJ, Dennis ES, Hemming MN, Trevaskis B: ODDSOC2 Is a MADS Box floral repressor that is down-regulated by vernalization in temperate cereals. Plant Physiol. 2010, 153: 1062-1073. 10.1104/pp.109.152488.PubMed CentralView ArticlePubMedGoogle Scholar
- Boyde A, Boyde S, Boyde A, Boyde S: Further studies of specimen volume changes during processing for SEM: including some plant tissue. Scan Electron Microsc. 1980, 2: 117-124.Google Scholar
- Moncur MW: Shrinkage of plant material during critical point drying. Scanning. 1979, 2: 175-177. 10.1002/sca.4950020309.View ArticleGoogle Scholar
- Talbot M, White R: Methanol fixation of plant tissue for scanning electron microscopy improves preservation of tissue morphology and dimensions. Plant Methods. 2013, 9: 36-10.1186/1746-4811-9-36. doi:10.1186/1746-4811-9-36PubMed CentralView ArticlePubMedGoogle Scholar
- Cameron IL, Hunter KE, Smith NKR: The subcellular concentration of ions and elements in thin cryosections of onion root meristem cells; an electron-probe EDS study. J Cell Sci. 1984, 72: 295-306.PubMedGoogle Scholar
- Whitehead DC, Goulden KM, Hartley RD: The distribution of nutrient elements in cell wall and other fractions of the herbage of some grasses and legumes. J Sci Food Agric. 1985, 36: 311-318. 10.1002/jsfa.2740360502.View ArticleGoogle Scholar
- Hepler PK: Calcium: a central regulator of plant growth and development. Plant Cell. 2005, 17: 2142-2155. 10.1105/tpc.105.032508.PubMed CentralView ArticlePubMedGoogle Scholar
- Fiji is just ImageJ.http://fiji.sc/Fiji,
- Schindelin J, Arganda-Carreras I, Frise E, Kaynig V, Longair M, Pietzsch T, Preibisch S, Rueden C, Saalfeld S, Scmid B, Tinevez J-Y, White DJ, Hartenstein V, Eliceiri K, Tomancak P, Cardona A: Fiji: an open-source platform for biological-image analysis. Nat Methods. 2012, 9 (7): 676-682. 10.1038/nmeth.2019.View ArticlePubMedGoogle Scholar
- Meijering E: Bioimage analysis website.http://www.imagescience.org/meijering/,
- Lichtenthaler HK, Schweiger J: Cell wall bound ferulic acid, the major substance of the blue-green fluorescence emission of plants. J Plant Physiol. 1998, 152: 272-282. 10.1016/S0176-1617(98)80142-9.View ArticleGoogle Scholar
- Joy D: A database on electron-solid interactions. 2008, Revision 08–1. [web.utk.edu/~srcutk/database.doc]Google Scholar
- Ager FJ, Ynsa MD, Dominguez-Solis JR, Gotor C, Respaldiza MA, Romero LC: Cadmium localization and quantification in the plant Arabidopsis thaliana using micro-PIXE. Nucl Inst Methods Phys Res B. 2002, 189: 494-498. 10.1016/S0168-583X(01)01130-2.View ArticleGoogle Scholar
- Grignon C, Sentenac H: pH and ionic conditions of the apoplast. Annu Rev Plant Physiol Plant Mol Biol. 1991, 42: 103-128. 10.1146/annurev.pp.42.060191.000535.View ArticleGoogle Scholar
- Haynes RJ: Ion exchange properties of roots and ionic interactions within the root apoplasm: their role in ion accumulation by plants. Bot Rev. 1980, 46 (1): 75-99. 10.1007/BF02860867.View ArticleGoogle Scholar
- Jarvis MC, Briggs SPH, Knox JP: Intercellular adhesion and cell separation in plants. Plant Cell Environ. 2003, 26: 977-989. 10.1046/j.1365-3040.2003.01034.x.View ArticleGoogle Scholar
- Morgan AJ: Preparation of specimens: changes in chemical integrity. Hayat MA: x-ray microanalysis in biology. 1981, London: Macmillan Publishers Ltd, 65-165.Google Scholar
- McCully ME, Canny MJ, Huang CX: Cryo-scanning electron microscopy (CSEM) in the advancement of functional plant biology. Morphological and anatomical applications. Funct Plant Biol. 2009, 36: 97-124. 10.1071/FP08304.View ArticleGoogle Scholar
- Chen W, Stoddard F, Baldwin TC: Variable pressure scanning electron microscopy of vicia faba stigmatic papillae. Microsc Anal. 2011, 25 (7): 17-18.Google Scholar
- Tang C-Y, Huang R-N, Kuo-Huang L-L, Kuo T-C, Yang Y-Y, Lin C-Y, Jane W-N, Chen S-J: A simple cryo-holder facilitates specimen observation under a conventional scanning electron microscope. Microsc Res Tech. 2012, 75: 103-111. 10.1002/jemt.21031.View ArticlePubMedGoogle Scholar
- Neinhuis C, Edelmann HG: Methanol as a rapid fixative for the investigation of plant surfaces for SEM. J Microsc. 1996, 184: 14-16. 10.1046/j.1365-2818.1996.d01-110.x.View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an open access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.