- Open Access
OSCILLATOR: A system for analysis of diurnal leaf growth using infrared photography combined with wavelet transformation
Plant Methodsvolume 8, Article number: 29 (2012)
Quantification of leaf movement is an important tool for characterising the effects of environmental signals and the circadian clock on plant development. Analysis of leaf movement is currently restricted by the attachment of sensors to the plant or dependent upon visible light for time-lapse photography. The study of leaf growth movement rhythms in mature plants under biological relevant conditions, e.g. diurnal light and dark conditions, is therefore problematic.
Here we present OSCILLATOR, an affordable system for the analysis of rhythmic leaf growth movement in mature plants. The system contains three modules: (1) Infrared time-lapse imaging of growing mature plants (2) measurement of projected distances between leaf tip and plant apex (leaf tip tracking growth-curves) and (3) extraction of phase, period and amplitude of leaf growth oscillations using wavelet analysis. A proof-of-principle is provided by characterising parameters of rhythmic leaf growth movement of different Arabidopsis thaliana accessions as well as of Petunia hybrida and Solanum lycopersicum plants under diurnal conditions. The amplitude of leaf oscillations correlated to published data on leaf angles, while amplitude and leaf length did not correlate, suggesting a distinct leaf growth profile for each accession. Arabidopsis mutant accession Landsberg erecta displayed a late phase (timing of peak oscillation) compared to other accessions and this trait appears unrelated to the ERECTA locus.
OSCILLATOR is a low cost and easy to implement system that can accurately and reproducibly quantify rhythmic growth of mature plants for different species under diurnal light/dark cycling.
The movement displayed by plants has long fascinated people and it is believed that ancient tribes used rhythmic leaf movements to schedule their prayers. The first documented attempt to elucidate whether the rhythm of movement was inherent to the plant or the result of external stimuli was performed by de Mairan in 1729. He observed that the rhythmic leaf movements of his ‘sensitive plant’ (Mimosa pudica) continued even in continuous darkness[1, 2]. Indeed, leaf movements of many species are controlled by the endogenous circadian clock and in the past decade plant circadian clock research has frequently used leaf movements of Arabidopsis thaliana seedlings as a marker of clock output which can be easily compared between different genotypes[4–6].
Analysis of rhythmic growth in seedlings
Various systems have been described for the analysis of leaf movement in Arabidopsis seedlings[4, 6]. These systems are characterised by sequential imaging of seedlings from the side. The position of the cotyledons or the first and second real leaf tip is than recovered from the time series images using for example NKTRACE, METAMORPH Software, or custom made programs. Subsequently, the quantified leaf position curves are often analysed using Biological Rhythms Analysis Software System (BRASS). Analysis of the leaf tip plots includes Fast Fourier Transform-NonLinear Least Squares (FFT-NLLS) analysis[9, 10]. FFT-NLLS provides the average phase and amplitude of cyclic processes, based on the best fitted sinusoidal curve over multiple days and thus does not capture the daily changes in phase and amplitude upon transition to a different growth condition or during development.
Analysis of rhythmic growth in mature plants
Methods developed for the analysis of circadian movements or upward leaf reorientation (hyponastic growth) in mature plants include physical attachment of sensors to the plant, photoelectric devices developed for measuring leaf movements in space independent of direct contact and strings attached to a rotation resistance transducer glued to the leaf. Time-lapse photography is another commonly used method (e.g.[8, 14, 15]). Imaging from the side makes it difficult to quantify leaf movements of mature plants as the dense whirl of (rosette) leaves may obscure a clear view of single leaves. For this reason, leaves obscuring the petiole base of the tracked leaf need to be removed. This procedure was previously used to quantify hyponastic growth in Arabidopsis and removal of leaves was reported not to influence the movement of the tracked leaves. Moreover, in order to correct for diurnal and circadian effects on leaf movement, the angles of treated and control plants were subtracted in these experiments. Another disadvantage of time-lapse photography is that it commonly depends on standard cameras, which require visible light. It is therefore only suitable for continuous light experiments. To simulate night conditions, non-photosynthetic green light was used to image Arabidopsis leaf growth during the dark period. Similarly infrared imaging has been used to measure the kinetics of Arabidopsis leaf reorientation in response to light quality, the response kinetics of Arabidopsis seedlings to ethylene and the growth rate of Arabidopsis hypocotyls in diurnal conditions. Alternatively, images taken at the beginning and end of the day period were used to analyse the diurnal leaf growth and movement of developing Arabidopsis rosette leaves. This approach gives an average growth rate for the light and dark period and can therefore not be used to determine phase, period or amplitude in the leaf growth movement. Here we developed a monitoring system based on infrared (IR) photography called OSCILLATOR. Our system allows continuous, high resolution growth analysis of mature rosette plants, independent of the presence of visible light. It thus enables measurement under biological relevant diurnal photoperiods. We positioned the IR sensitive cameras above the plants. Although imaging from above does not always allow for exact quantification of leaf length due to leaf hyponasty, a top view avoids the problem of rosette leaves obscuring each other. In addition it still allows reliable extraction of leaf movement parameters (phase, period, amplitude) in model species such as Arabidopsis, Petunia hybrida (petunia) and Solanum lycopersicum L. (tomato) plants. By tracking the movement of the leaf tip of specific leaves over time (typically seven days), growth movement curves were obtained from which phase and amplitude were extracted using wavelet analysis. This processing method allows for reliable measurement of daily phase and amplitude which are convenient parameters to quantify the effect of mutant genes or physiological treatments on growth. To validate the system, we determined the natural variation for diurnal leaf growth movement in several Arabidopsis accessions.
Results and discussion
The OSCILLATOR system for continuous analysis of plant growth under continuing diurnal light/dark cycles consists of three modules: (1) data acquisition in the experimental setup (2) image processing and (3) extraction of phase, period and amplitude using wavelet analysis (see Figure1).
Data acquisition in the experimental setup
The hardware of the system consists of a climate controlled growth cabinet fitted with two IR LED light units (890 nm) and two modified single-lens reflex (SLR) cameras with the IR filter removed. The cameras are fitted to a sliding frame to allow easy positioning above the plants (Figure1a). With two cameras per cabinet the full surface area of the growth cabinet could be monitored (Figure1b) with minimal ‘visual angle-to-object’ effects. Cameras are controlled using NIKON Camera Control software on dedicated laptops. Arabidopsis rosette plants (32 days old) are placed on an irrigation mat which is saturated with tap water every three days. After seven days of acclimation in the growth cabinet images are taken every twenty minutes over a period of up to 16 days.
From the obtained digital image stack of multiple plants, single plant frames were cropped using ImageJ freeware (Figure1c). For Arabidopsis, for each individual plant, the 11th and 12th real leaves (5 ~ 7 mm long at t = 0) were selected in the first frame and used for measurement of the projected distance between leaf tip to rosette centre throughout the image stack (Figure1c). This can be achieved in two ways: 1: ImageJ “manual tracking” plugin allows manual tracking of the leaf tip position. 2: ImageJ “Particle tracker” allows measurement of the position of a small dot of inert paint placed on the leaf tip at the start of imaging (this did not affect growth). Both methods provide Microsoft Excel compatible files containing the leaf tip coordinates in pixels (Figure1c). The rosette centre coordinates were similarly determined in the first frame. For Arabidopsis these remained fixed throughout the experiment. The projected distance in pixels between leaf tip and rosette centre was calculated and converted into millimetres according to a scale marker (placed at average plant height) included in the images. This distance is plotted against time and represents projected leaf length (Figure1d).
Extraction of phase, period and amplitude using wavelet analysis
In the projected length curve the vertical rhythmic leaf movements are identified as oscillations in the curve. For the extraction of these oscillations the individualcurves are imported into R freeware and fitted with a best fit 2° polynomial regression line representing average projected growth rate (R2 > 0.85) (Figure1d). The regression curve was subtracted from the leaf tip movement curve, providing a residual oscillation curve. Decreasing distance between tip and centre indicates upward leaf growth movement. Because we want to use the maximum hyponastic leaf position as amarker for the phase of leaf movement, the residual oscillations were inverted to allow maximum upright leaf position to correspond to maximum peak height (Compare Figure1d and1e). We confirmed that the peak of oscillations indeed corresponds to the highest leaf angle: For one set of plants the absolute lengths of the tracked leaves were measured daily at the end of the photoperiod (Additional file1: Figure S1a). Based on the measured projected and absolute leaf lengths, for each day the leaf angle was calculated. A comparison between the smoothed projected leaf tip oscillations and the calculated leaf angle confirmed that peak oscillations match maximal leaf angles (Additional file1: Figure S1b). On the transition from day to night period a ‘bump’ is observed in the projected leaf growth oscillations. This is caused by a temporary relapse in the upward leaf movement on the light to dark transition (Figure1e). In order to obtain an objective phase and amplitude of leaf movement, the raw projected oscillation plots of individual leaves were smoothed by removal of harmonic noise using wavelet analysis based on WAVECLOCK (Figure1f). Wavelet analysis provides an alternative for the commonly used FFT-NLLS method and allows for an accurate day-to-day estimation of phase and amplitude. With wavelet analysis we therefore get an accurate description of adaptations in phase and amplitude throughout development. Smoothed projected oscillation curves of 10 to 12 leaves from 5 to 6 individual plants were averaged and plotted with standard errors for each time point (Figure1f).
Characteristics of leaf growth movement in Arabidopsis rosette plants
OSCILLATOR was used to characterise growth of 32 day old Arabidopsis Col-0 rosette plants. For each individual plant, the 11th and 12th real leaves (5 ~ 7 mm long at t = 0) were analysed for seven days. During this period, the projected lengths increased from ~7 to ~ 42 mm. Oscillations initially increased with progressing development, but decreased after ~144 hours and leaves were no longer moving after nine days (Additional file1: Figure S1c). For characterisation of the phase and amplitude of leaf growth movement in subsequent experiments we therefore chose the developmental window of seven days during which robust oscillations were observed (Additional file1: Figure S1c). It was previously shown that low levels of IR illumination did not influence seedling development. Nevertheless, we compared projected leaf lengths between plants grown with IR lights (allowing night measurements) and plants grown without IR (day only measurements). Results show an overlap between the IR and non-IR day measurements and final projected leaf length did not differ under these two conditions (Additional file1: Figure S1a). Therefore, we conclude that also in our system the supplemental IR light does not influence leaf growth movement.
Natural variation in leaf growth movements
Previously, natural variation in the angle of the petiole of Arabidopsis accessions was determined at a fixed time of the day. In this assay the petiole angle of the different accessions varied between 15.3 degrees in Warschau-1 (Wa-1) to 52.0 degrees in Meloy Ornes. For characterisation of the leaf growth movement we selected six accessions with varying petiole angles, including the common ‘laboratory accessions’ Col-0, Ler-1 and Ws-2 (Additional file2: Figure S2). For each accession the projected leaf lengths and extracted leaf growth oscillations were determined (Figure2). Results show distinct differences between leaf movement amplitude in the different accessions. Average amplitudes varied among the dataset from 1.89 ± 0.23 mm (Wa-1) to 6.83 ± 0.56 mm (Ts-1) (Figure2h,j, Figure3a). To verify whether variation in leaf petiole angle relates to leaf movement amplitude under diurnal conditions the correlation between reported leaf angles and the amplitudes obtained through OSCILLATOR was determined. Indeed the correlation between amplitude of leaf movement and initial petiole angles was very strong (R2 = 0.8756) (Figure3b). Differences in leaf length in the different accessions could contribute to differences in the measured amplitude. To examine the correlation between leaf length and amplitude these parameters were compared for all accessions. However, no significant correlation between the two was found (Additional file1: Figure S1d). This indicates that variation in amplitude does not result from variation in leaf length and each accession has a distinct leaf growth profile. The period of leaf movement did not differ significantly and was ~24 hours for all accessions, which may be expected under the diurnal entrainment regime (Figure3c). The phase of leaf movement varied somewhat in time but was most stable between two and six days where the strongest amplitudes were observed (Additional file3: Figure S3). The average phase for this period differed between the accessions (P = 0.018). The phase of Ler-1 (16.9 hours) was significantly later than that of Ts-1 (14.5 hr., P = 0.016), Ws-2 (15.2 hr., P = 0.039) and Wa-1 (15.7 hr., P = 0.048) (Figure3d).
The late phase of Ler-1 leaf oscillations appears unrelated to the ERECTA locus
We investigated whether the late phase of the Ler-1 relates to the null allele of the ERECTA (ER) locus in this accession. ER encodes a leucine-rich repeat receptor-like Ser/Thr kinase, and Ler carries a missense mutation within the conserved region of the kinase domain. Therefore, ER activity differs between Landsberg erecta and the original Landsberg wild-type line (Lan-0) from which Ler was isolated. ER has previously been reported to control ethylene induced leaf hyponasty[25, 26]. However, it was not investigated whether ER affects the phase of leaf movement. Leaf oscillations of Ler-1 were compared to those of Lan-0 (Figure4a) and results show that the observed late phase of Ler-1 under diurnal light and temperature cycles is also present in Lan-0 (Figure4b-c). This indicates that in both accessions genetic variation independent of ER is responsible for this phenotype.
Leaf growth movement in petunia and tomato plants
The OSCILLATOR system was developed and optimised for Arabidopsis. To test if the system can be used for other species without major modification 32 day old plants of two additional model species; petunia(Figure5a) and tomato (Figure5b) were analysed. In its vegetative stage petunia has a rosette structure (Figure5a) and therefore growth analysis could be measured using OSCILLATOR without any modification. Figure5c shows the projected leaf length measured for the petunia plants. From these curves the projected oscillations were extracted (Figure5d). Tomato plants displayed strong circumnutations (variable apex position in time). This made tracking of the central meristem necessary for calculation of the projected distance. After correction for centre displacement, clear diurnal rhythms in leaf growth movement could be extracted (Figure5d).Both petunia (V26) and tomato (Money Maker) displayed a phase of 18.0 ± 0.27 and 18.3 ± 1.21 hours respectively, which is about two hours later than the phase of Arabidopsis (e.g. 16.1 ± 0.21 hr. for Col-0). Results indicate that each species displays a unique leaf growth movement pattern which can be considered the integrated result of the effects of light, temperature and the endogenous circadian oscillator on plant development.
The natural variation we identified within six Arabidopsis accessions matched previous described observations, demonstrating the validity of our system. Variation among natural accessions has been studied under continuous light conditions before[5, 27]. OSCILLATOR now provides the possibility to study leaf growth and movement under various diurnal conditions, which more closely mimic the natural environment. Characterisation of leaf growth movement by phase, period and amplitude allows easy comparison between different genotypes and treatments. Our system also opens up the perhaps even more intriguing possibility to evaluate plant behaviour under gradual changing photoperiods (mimicking seasonal transitions). Diurnal rhythms in leaf growth and movement are directly related to plant growth and help repositioning of leaves relative to the light and could contribute to increased photosynthetic capacities[22, 28, 29]. Furthermore, leaf growth movement and related changes in rosette compactness have been shown to facilitate cooling and allow adaptation to increasing ambient temperatures. Combined, the results show that OSCILLATOR can be used to extract parameters of leaf growth movement which can be used to characterise different genotypes. OSCILLATOR provides plant scientists with a relatively cheap, reliable and non-invasive tool to accurately dissect diurnal growth rhythms of various plant species under continuing day/night cycles.
Plant material and growth conditions
Seeds of Arabidopsis thaliana accessions were provided by M. van Zanten (Laboratory of Molecular Plant Physiology, Utrecht University, The Netherlands) and J. Keurentjes (Laboratory of Plant Genetics, Wageningen University, The Netherlands). Seeds of Petunia hybrida (v26) were donated by Tom Gerats (Laboratory of Plant Genetics, Radboud University Nijmegen, The Netherlands). Seeds of tomato (S. lycopersicum L.) cultivar; Money Maker were obtained from Wouter Kohlen (Max Planck Institute for Plant Breeding Research, Köln, Germany). All experiments were performed in automated climate controlled WEISS (http://www.wkt.com) cabinets (12/12 hours light/dark cycle). Relative humidity was kept constant at 60% (v/v) and photosynthetic active radiation (PAR) was 150 μmol m2 s-1 from white fluorescents tubes (PHILIPS, type T5, Colour code: 840). Ambient temperature cycles for growth were 22°C (photoperiod) and 12°C (dark period) with a temperature ramp of 0.33°C/min. Measurements showed that soil temperature lagged ~20 minutes behind ambient air temperature. Plants were grown in fertilised peat / perlite based soil in square (5x5x5 cm) plastic pots with different genotypes placed at random positions in the growth cabinet. Plants were placed on an irrigation mat which was watered automatically to saturation through porous tubing from a basin containing tap-water every three days at the start of the photoperiod. After 20 days plants were watered once with half strength Hoagland-nutrient solution instead of water. Five days later plants were transferred to a second climate cabinet for imaging with similar conditions and an IR camera system with IR lights. Plants were allowed to acclimate for seven days before the onset of imaging. Thus, at the start of imaging plants were 32 days old and the Arabidopsis accessions all had 13–14 true leaves.
Plant growth imaging and image data analysis
The pipeline for imaging and image analysis as used by OSCILLATOR is summarised in figure S4. To enable automated leaf tip tracking the 11th and 12th real leaf was marked with inert paint before the start of imaging although for manual tip tracking this is not necessary (tip tracking will be explained in more detail later). The 11th and 12th leaf were analysed and the leaf length varied from 5–8 mm. Imaging was with SLR NIKON D90 digital camera’s with a NIKON AF 50 mm F/1.8 lens. Cameras were powered by net adaptors (http://www.nikon.com). To enable night photography, cameras were custom modified by MAXMAX (http://www.maxmax.com) for removal of the internal IR filter to allow detection of IR light. Sufficient IR illumination per cabinet was provided by two continuous burning LED lights (890 nm, 12 W, KÖNIG electronics, (http://www.konigelectronic.com). Each camera was connected to a dedicated laptop with active USB 2.0 repeater cables and controlled with time-lapse photography software (NIKON Camera Control Pro 2;http://www.nikon.com). Camera settings were fixed; F/stop = f/8, Exposure time = 1/5 sec., ISO speed = ISO-200. Field of view for each camera was 16 rosette plants. Photographs were taken every 20 minutes and stored as individual images. Sequential images were imported as virtual stacks into ImageJ (http://rsbweb.nih.gov/ij/) and the image stack was subsequently cropped into individual plant image stacks. For cropping of single plant image stacks, the desired areas can be selected in the first frame and cropping of all the stacks is further automatic (Additional file4: Figure S4b). The resulting multiple single plant ‘virtual stacks’ were manually saved as individual plant image stacks and named as appropriate. Each individual plant image stack was used for leaf tip tracking, using either the manual tracking (Additional file5: File S1,http://rsbweb.nih.gov/ij/plugins/track/track.html) or the automated MOSAIC particle tracking plugin (Additional file5: File S1,http://www.mosaic.ethz.ch/Downloads/ParticleTracker). The manual tracking plugin allows semi-automated selection of leaf tip coordinates throughout the stack (Additional file4: Figure S4c). To facilitate this procedure Standard Mouse Auto Clicker 2.8 (Additional file5: File S1) can be used to automate screen mouse clicks at specified intervals and any location on the screen. Alternatively, when selected leaves are marked with a small paint dot at the start of the experiment, automated tracking of the leaf tip with the particle tracker MOZAIC plugin can be used. However, the particle recognition occasionally fails in single frames, resulting in gaps in the leaf tip tracks. This then requires manual correction, which can be labour intensive. Therefore, in this work we used the manual tracking plugin. In combination with Standard Mouse Auto Clicker 2.8 (Additional file5: File S1) set at 1 click per 0.2 second a typical stack of 500 images is manually processed in 100 seconds. Both the manual and automated tracking plugins provide output format compatible with MICROSOFT EXCEL 2010 (http://office.microsoft.com) which can be individually named and saved by the user for each individual leaf track (e.g. Plant1.1.xls etc.; Additional file4: Figure S4d).
Extraction of parameters of leaf growth oscillations with the OSCILLATOR script
To determine the relative leaf movement, for each image the distance between leaf tip and plant centre needs to be calculated. In our experiments the Arabidopsis and petunia rosette centres were static throughout the image stack and thus were provided by the single manually determined (x/y) coordinates of the rosette apical centre. For tomato the apex positions varied throughout the stack and apex position was determined using ‘manual tracking plugin’ (Additional file5: File S1). The leaf tip and apex coordinates of all individual leaves of a single genotype (2 leaves per plant, 4 plants per genotype) are subsequently combined in the OSCILLATOR_input.csv file (Additional file5: File S1, Additional file4: Figure S4e). For each genotype, the OSCILLATOR_input.csv file was placed in designated folders each containing the OSCILLATOR SCRIPT (Additional file5: File S1, Additional file4: Figure S4f). For each folder the script was activated in R 2.13.1 (http://www.r-project.org/), which generates the following outputs:
Projected length for individual leaves and averaged projected length
The distance between the leaf tip and the plant apex was calculated by the OSCILLATOR.R script based on the following equation:
The resulting distance (projected leaf length) is provided for single leaves in output file project_length.csv. In addition the average is plotted against time as a graph with SE depicted as shading (Projectedlength.jpeg).
Raw projected oscillations for individual leaves and averaged raw projected oscillations
Subsequently a best fit second degree polynomial trend line is automatically calculated for each individual projected length curve and the residual values are subtracted from this line. The resulting residuals were inverted to allow maximum upright leaf position to correspond to maximum peak height. These raw projected oscillations are provided as data in output file AvgRawOscillation.csv. In addition the average is plotted against time with SE as shading in AvgRawOscillations.jpeg (Additional file4: Figure S4g).
Smoothed projected oscillations for individual leaves and smoothed averaged projected oscillations
The raw oscillations of individual leaves are then smoothed using WAVECLOCK script imbedded in the OSCILLATOR script (Additional file5: File S1). The smoothed projected oscillations are provided for single leaves in the file IndividualSmoothedOscillation.csv and as an average with SE in the file AverageSmoothedOscillations.csv. In addition the average is plotted against time with SE depicted as shading in AvgSmoothedOscillation.jpeg (Additional file4: Figure S4g).
Phase and amplitude of individual leaves and averaged phase, amplitudes and period
The peak values for the smoothed projected oscillations of individual leaves for all periods are used to calculate the phase and amplitude. The phase and amplitude data are provided for single leaves in file phaseDays.csv and amplitudeDays.csv respectively and are then averaged and plotted against time with SE in graph RplotphaseMinMax.jpeg and RplotampMinMax.jpeg respectively (Additional file4: Figure S4g). Period information is plotted as a wavelet scalogram (Wavelet_1.png).
Given all materials and software in place it will take an experienced user approximately one hour to complete a single genotype set (four plants, eight leaves) analysed throughout a stack of imaging representing seven days of development as typically performed in this study. All data presented is the typical result of at least two independent experiments, each based on at least 7 individual leaf track analysis.
Statistical differences between accessions were determined using one-way ANOVA. Individual differences were then identified using a post-hoc Tukey test (P < 0.05). All analysis were performed using SAS_9.20 (http://www.sas.com/).
Mancuso S, Shabala S, Moran N: Rhythmic Leaf Movements: Physiological and Molecular Aspects. Rhythms in Plants. 2007, Berlin Heidelberg: Springer, 3-37.
de Mairan J: Observation botanique. Hist Acad Roy Sci. 1729, 35-36.
McClung CR: Plant circadian rhythms. Plant Cell. 2006, 18: 792-803. 10.1105/tpc.106.040980.
Engelmann W, Simon K, Phen CJ: Leaf movement rhythms in Arabidopsis thaliana. Z Naturforsch. 1992, 47: 925-928.
Swarup K, Alonso-Blanco C, Lynn JR, Michaels SD, Amasino RM, Koornneef M, Millar AJ: Natural allelic variation identifies new genes in the Arabidopsis circadian system. Plant J. 1999, 20: 67-77. 10.1046/j.1365-313X.1999.00577.x.
Edwards KD, Millar AJ: Circadian Rhythms : Methods and Protocols. 2007, 103-113. Analysis of circadian leaf movement rhythms inArabidopsis thaliana, Methods in Molecular Biology
Onai K, Okamoto K, Nishimoto H, Morioka C, Hirano M, Kami-ike N, Ishiura M: Large-scale screening of Arabidopsis circadian clock mutants by a high-throughput real-time bioluminescence monitoring system. Plant J. 2004, 40: 1-11. 10.1111/j.1365-313X.2004.02191.x.
Benschop JJ, Millenaar FF, Smeets ME, van Zanten M, Voesenek LA, Peeters AJ: Abscisic acid antagonizes ethylene-induced hyponastic growth in Arabidopsis. Plant Physiol. 2007, 143: 1013-1023.
Plautz JD, Straume M, Stanewsky R, Jamison CF, Brandes C, Dowse HB, Hall JC, Kay SA: Quantitative analysis of Drosophila period gene transcription in living animals. J Biol Rhythms. 1997, 12: 204-217. 10.1177/074873049701200302.
Dowson-Day MJ, Millar AJ: Circadian dysfunction causes aberrant hypocotyl elongation patterns in Arabidopsis. Plant J. 1999, 17: 63-71. 10.1046/j.1365-313X.1999.00353.x.
Halaban R: The Circadian Rhythm of Leaf Movement of Coleus blumei x C. frederici, a Short Day Plant. II. The Effects of Light and Temperature Signals. Plant Physiol. 1968, 43: 1887-1893. 10.1104/pp.43.12.1887.
Tibbitts TW, Hoshizaki T, Alford DK: Photoelectric device for recording of leaf movements. Orig Life Evol B. 1970, 2: 109-112. 10.1007/BF00928958.
Poiré R, Wiese-Klinkenberg A, Parent B, Mielewczik M, Schurr U, Tardieu F, Walter A: Diel time-courses of leaf growth in monocot and dicot species: endogenous rhythms and temperature effects. J Exp Bot. 2010, 61: 1751-1759. 10.1093/jxb/erq049.
Granier C, Aguirrezabal L, Chenu K, Cookson SJ, Dauzat M, Hamard P, Thioux J-J, Rolland G, Bouchier-Combaud S, Lebaudy A: PHENOPSIS, an automated platform for reproducible phenotyping of plant responses to soil water deficit in Arabidopsis thaliana permitted the identification of an accession with low sensitivity to soil water deficit. New Phytol. 2006, 169: 623-635. 10.1111/j.1469-8137.2005.01609.x.
Mullen JL, Weinig C, Hangarter RP: Shade avoidance and the regulation of leaf inclination in Arabidopsis. Plant Cell Environ. 2006, 29: 1099-1106. 10.1111/j.1365-3040.2005.01484.x.
Zhang X, Hause RJ, Borevitz JO: Natural Genetic Variation for Growth and Development Revealed by High-Throughput Phenotyping in Arabidopsis thaliana. Genes Genomes Genet. 2012, 2: 29-34.
Binder BM, O'Malley RC, Wang W, Moore JM, Parks BM, Spalding EP, Bleecker AB: Arabidopsis seedling growth response and recovery to ethylene. A kinetic analysis. Plant Physiol. 2004, 136: 2913-2920. 10.1104/pp.104.050369.
Nozue K, Covington MF, Duek PD, Lorrain S, Fankhauser C, Harmer SL, Maloof JN: Rhythmic growth explained by coincidence between internal and external cues. Nature. 2007, 448: 358-361. 10.1038/nature05946.
Pantin F, Simonneau T, Rolland G, Dauzat M, Muller B: Control of Leaf Expansion: A Developmental Switch from Metabolics to Hydraulics. Plant Physiol. 2011, 156: 803-815. 10.1104/pp.111.176289.
Sbalzarini IF, Koumoutsakos P: Feature point tracking and trajectory analysis for video imaging in cell biology. J Struct Biol. 2005, 151: 182-195. 10.1016/j.jsb.2005.06.002.
Price TS, Baggs JE, Curtis AM, FitzGerald GA, Hogenesch JB: WAVECLOCK: wavelet analysis of circadian oscillation. BMC. 2008, 24: 2794-2795.
van Zanten M, Pons TL, Janssen JAM, Voesenek LACJ, Peeters AJM: On the Relevance and Control of Leaf Angle. Crit Rev Plant Sci. 2010, 29: 300-316. 10.1080/07352689.2010.502086.
Torii KU, Mitsukawa N, Oosumi T, Matsuura Y, Yokoyama R, Whittier RF, Komeda Y: The Arabidopsis ERECTA gene encodes a putative receptor protein kinase with extracellular leucine-rich repeats. Plant Cell. 1996, 8: 735-746.
Redei GP: Supervital Mutants of Arabidopsis. Genetics. 1962, 47: 443-460.
Millenaar FF, Cox MCH, van Berkel YEMD, Welschen RAM, Pierik R, Voesenek LAJC, Peeters AJM: Ethylene-induced differential growth of petioles in arabidopsis. Analyzing natural variation, response kinetics, and regulation. Plant Physiol. 2005, 137: 998-1008. 10.1104/pp.104.053967.
van Zanten M, Basten Snoek L, van Eck-Stouten E, Proveniers MC, Torii KU, Voesenek LA, Peeters AJ, Millenaar FF: Ethylene-induced hyponastic growth in Arabidopsis thaliana is controlled by ERECTA. Plant J. 2010, 61: 83-95. 10.1111/j.1365-313X.2009.04035.x.
Michael TP, Salomé PA, Yu HJ, Spencer TR, Sharp EL, McPeek MA, Alonso JM, Ecker JR, McClung CR: Enhanced Fitness Conferred by Naturally Occurring Variation in the Circadian Clock. Science. 2003, 302: 1049-1053. 10.1126/science.1082971.
Abreu ME, Munné-Bosch S: Hyponastic leaf growth decreases the photoprotective demand, prevents damage to photosystem II and delays leaf senescence in Salvia broussonetii plants. Physiol Plant. 2008, 134: 369-379. 10.1111/j.1399-3054.2008.01144.x.
Hopkins R, Schmitt J, Stinchcombe JR: A latitudinal cline and response to vernalization in leaf angle and morphology in Arabidopsis thaliana (Brassicaceae). New Phytol. 2008, 179: 155-164. 10.1111/j.1469-8137.2008.02447.x.
Crawford AJ, MD H, Alistair M, Franklin KA: High temperature exposure increases plant cooling capacity. Curr Biol. 2012, 22: 396-397. 10.1016/j.cub.2012.03.044.
This research is funded by the Top Technological Institute Green Genetics (TTI-GG) (Project: 2CFL009RP). We are indebted to: R. Pierik (Utrecht University), M. van Zanten (Utrecht University), J. Rienks (Wageningen University), N. Zagari (Wageningen University) and W. Kohlen (Max-Planck institute, Köln) for providing essential feedback on the method and manuscript. We thank T. Stoker and G. Stunnenberg (Wageningen University) for maintaining the Weiss cabinets.
The authors declare that they have no competing interests.
The experiments were planned by RB and AvdK. RB performed the experiments, analysed the data and prepared the figures. MM constructed the OSCILLATOR. R script. The manuscript was written by RB, AvdK and HB. All authors read and approved the final manuscript.