- Open Access
A petal breakstrength meter for Arabidopsis abscission studies
© Lease et al; licensee BioMed Central Ltd. 2006
Received: 06 January 2006
Accepted: 16 February 2006
Published: 16 February 2006
Abscission is the regulated dropping of plant organs, such as leaves or flower petals. This process involves a break down of the cell wall between layers of cells in the abscission zone, causing the organ to become detached. The model plant Arabidopsis thaliana undergoes floral organ abscission. Various experimental methods have been used to study Arabidopsis floral organ abscission, including measuring the petal breakstrength, or the amount of force required to pull a petal from the receptacle. Petal breakstrength provides a quantitative insight into the physical integrity of the petal abscission zone.
We developed a petal breakstrength meter that allows rapid data acquisition on a personal computer. We present the design of the device and show its utility in measuring Arabidopsis petal breakstrength for abscission studies.
This petal breakstrength meter should enable researchers to perform the petal breakstrength assay as a routine part of the characterization of environmental and genetic factors affecting abscission.
The phenomenon of trees dropping their leaves in autumn is an example of abscission familiar to most people. Abscission, the regulated shedding of plant organs, can involve loss of other plant organs besides leaves, such as flower petals. At the cellular level, abscission involves a separation between layers of cells in the abscission zone, allowing a plant organ to detach [1, 2]. For the separation to occur in a controlled fashion, there must be regulated expression of the enzymes involved in breaking down the components of the cell wall in the middle lamella.
A powerful method for identifying the regulatory components of abscission has been the study of Arabidopsis thaliana mutants that affect this process. Unlike many plants, Arabidopsis does not shed its leaves. However, Arabidopsis does undergo floral organ abscission, such as petal dropping. By identifying mutants with altered abscission, gene products and pathways important in abscission have been uncovered. For example, ethylene perception has been shown to be important in this response . Furthermore, ACTIN-RELATED PROTEIN7 (ARP7) knockdowns exhibited delayed floral abscission . Moreover, a cell surface receptor involved in abscission is HAESA, a leucine-rich repeat receptor kinase . Inflorescence Deficient in Abscission (IDA), encodes a small secreted protein that is necessary for abscission . BLADE-ON-PETIOLE1 and BLADE-ON-PETIOLE2 genes affect floral organ abscission . Overexpression of AGL15 causes abscission defects (Fernandez et al., 2000). Aditionally, AUXIN RESPONSE FACTOR1 and AUXIN RESPONSE FACTOR2 mutants also affect abscission [8, 9].
To understand the mechanistic basis of the abscission defects in these Arabidopsis mutants, several experimental approaches have been used. For example, the abscission zone has been visualized by microscopy . In addition, the expression of a molecular reporter associated with abscission zones has been investigated . Another assay that has been used in a subset of Arabidopsis abscission studies is the petal breakstrength assay [10, 6, 3]. In this assay, the force required to pull a petal from the receptacle is measured. This assay gives quantitative insight into the physical integrity of the abscission zone. For example, if the abscission zone is defective, more force may be required to pull the petal off than if abscission has been activated.
Part of the reason this assay has only been employed in a subset of Arabidopsis abscission studies may be that the device required to perform the assay is not a common piece of laboratory equipment. Furthermore, there is no off-the-shelf product that can be purchased to perform the assay, instead it must be built by the investigator. A petal breakstrength meter has been briefly described by Patterson and Bleecker (2004), which consisted of a FORT10 stress transducer (World Precision Instruments, Sarasota, FL) and a voltmeter. Further details of the device have only been published in a doctoral dissertation chapter , and so are not widely available. When we sought to build a petal breakstrength meter, we initially sought to purchase the FORT10 transducer, but we found that the device was no longer in production and was unavailable. Therefore, we designed the petal breakstrength meter around another force sensor which is in production, the MLT050 (ADInstruments, Colorado Springs, CO). We gave the petal breakstrength meter an additional feature: high speed data acquisition and logging to a personal computer. The design and use of our petal breakstrength meter should enable other researchers to build the meter for their abscission studies.
Basic operation of the device in the petal breakstrength assay
Explanation of the circuit
Improving sensitivity of MLT050 sensor to small forces
The MLT050 force transducer has a range of 0 – 50 grams and outputs a voltage signal that varies linearly with the applied force. However, since the voltage output of the sensor is so small (at 5 V device excitation, 50 g of applied force causes an output of less than 100 mV), directly measuring the voltage output of the sensor is not ideal. Direct measurement of the output of the sensor would prevent detecting subtle differences in petal breakstrength that may exist between abscission defect mutants.
Arabidopsis petal breakstrength requires several grams or less of force equivalents (Butenko et al., 2003). To improve the sensitivity of the meter to this range of measurement, we introduced two design features. First, we used mechanical leverage to improve sensitivity. A 5 cm long aluminum strip in which small holes were drilled at both ends was employed as a lever (Fig. 1B). The strip was fastened to the sensor with a screw and nut. Second, we linearly amplified the voltage signal output from the sensor by a factor of 100 using an operational amplifier (Fig. 2). These elements enabled the meter to have higher sensitivity to small forces and higher accuracy in the force range of interest.
Sensor attachment to Arabidopsis petals via petal gripper
Due to the miniscule size of Arabidopsis flowers, physically attaching the sensor to a flower petal was a challenging design issue. We initially tried micro-alligator clips, but found that they were too large and cumbersome. Instead, we used a clip that is designed for attaching to the legs of a surface mount device (SMD) integrated chip, called an 'SMD grabber.' The tips of the off-the-shelf SMD grabber were found to be too sharp and fine. As a consequence, during the assay, the flower petal would tear rather than be pulled out of the flower. We superglued ~1 mm long pieces of insulation from 20 gauge wire over the tips of the SMD grabber (Fig. 1C–E). This modification spreads out the pressure of the SMD grabber tips to a wider surface of the flower petal, and solved the petal tearing problem. The petal gripper is coupled to the sensor via a short piece of 20 gauge braided copper wire passed through the SMD grabber plunger and the drilled hole on the distal end of the 5 cm aluminum strip (Fig. 1B).
Device validation: linearity of response
Device validation in the petal breakstrength assay
To show the high resolution of data acqusition and visualize the data from an assay recording, we plotted the voltage measurements from a typical petal breakstrength trial (Figure 4B). The rate of data acquisition is greater than 100 measurements per second. During the petal breakstrength assay, the voltage is observed to rise as the petal is pulled until the moment the petal detaches from the receptacle. Following the petal pulling, the voltage briefly fluctuates as the lever vibrates with a harmonic motion before settling down to basal levels.
We tested our petal breakstrength meter in the petal breakstrength assay to see if the values we measured were comparable to those reported in the literature (Fig. 4C). For example, our measured average value of 2.24 grams for position two flower petals is close to the range of force (~1.75 g, ~2.5 g and ~1.75 g respectively) reported by Fernandez et al. (2000), Butenko et al. (2003), and Patterson and Bleecker (2004) in wildtype petal breakstrength assays. Therefore, we concluded that our device for petal breakstrength assay performed similarly to the preexisting device.
We have demonstrated the design and use of a petal breakstrength meter for Arabidopsis abscission studies. The detailed description of this device should enable other researchers to build the meter so that measuring petal abscission breakstrength will become a routine assay in studying new mutants. Excluding the personal computer, the cost of the parts used in construction (Table 1), is about $300 with the sensor being the most expensive single part. Assembling the device requires a soldering iron and about two hours of time. Therefore, the cost and time required for assembly of the petal breakstrength meter are not prohibitively high.
We have considered several future technical improvements of the device. For example, using an external 16 bit ADC instead of the onboard 10 bit ADC on the PIC18F452 would enable higher resolution measurements. In terms of speed of data acquisition, the serial transmission of data is the rate-limiting step. Two methods could be employed to improve this speed. First, a higher speed, e.g. 20 MHz crystal could be used for clocking the PIC, so that a higher rate of serial communication could be used reliably. Alternatively, the device could be modified to send the voltage measurement as two bytes of binary data, rather than a multibyte string of ASCII characters representing the decimal version of the data. This change would be rather easy to include as it would only involve modifying the firmware of the chip, rather than modifying the circuit hardware.
Parts and Suppliers for Construction of Petal Breakstrength Meter
(1) MLT050/D Force Transducer (ADInstruments, http://www.adinstruments.com)
(1) NJM2132D (Mouser, http://www.mouser.com, p\n 513-NJM2132D)
(1) SMD grabber clip (Mouser, p\n 565-5243-0)
(2) 16 pin DIP sockets (Mouser, p\n 575-199316)
(1) PIC18F452-I/P (Glitchbuster, http://www.glitchbuster.com p\n PIC18F452-I/P).
(1) 7805 voltage regulator (Glitchbuster, p\n 7805)
(1) MAX232N (Glitchbuster, p\n MAX232N)
(1) 4 MHz crystal (Glitchbuster, p\n XT-4)
(2) 22 pf capacitors (Glitchbuster, included with crystal)
(1) 4.7K ohm 0.25W resistor (Glitchbuster, p\n R4.7k)
(1) 470 ohm resistor 0.25W (Glitchbuster, p\n R470)
(5)10K ohm resistors (Glitchbuster, p\n R10K)
(2) 1.0 Mohm W resistors (Glitchbuster, p\n R1.0M)
(3) 100 Kohm W resistors (Glitchbuster, p\n R100k)
(4) 1.0 uF electrolytic capacitors (Glitchbuster, p\n 1R50)
(1) 4.7 uF electrolytic capacitor (Glitchbuster, p\n 4.7R50)
(1) 0.1 uF capacitor (Glitchbuster, p\n .1UF-MONO)
(1) Red LED (Glitchbuster, p\n LED-R)
(1) Serial cable (Glitchbuster, p\n DB9-6 M-F)
(1) STD DIN-8 female connector (MPJA, http://www.mpja.com, p\n 5616 PL)
(1) D subminiature 9 female right angle connector (MPJA, p\n 4840 PL)
(1) SPST switch (MPJA, p\n 5002 SW)
(1) ABS project enclosure (MPJA, p\n 15521 BX)
(1) Power supply 9V 1A or similar (MPJA, p\n12254 PD)
(1) perforated board (Radio Shack, p\n 276-147)
(1) 40 pin socket (Radio Shack, http://www.radioshack.com, p\n 276-1996)
We developed and tested a new petal breakstrength meter that provides a significant advance in the petal breakstrength assay used in Arabidopsis abscission studies. This should enable researchers to make this assay a routine part of their characterization of abscission mutants. In addition, the device may find unanticipated applications in scientific assays involving measurements of the physical integrity of other materials.
The electronic parts were obtained from the suppliers listed in Table 1. This list is given for convenience and should not be interpreted to read that the parts can only be obtained from the stated sources (with the exception of the force transducer). Multiple alternate vendors of the parts can be found by searching the internet.
In the event that the PIC18F452 is unavailable, the PIC18F4520 may be used as an alternative without modifying the circuit or Picbasic source code. The programmed device may be obtained by request from the corresponding author. Alternatively, the programmed PIC may be purchased from a commercial PIC programming service provider (e.g. http://www.glitchbuster.com).
The circuit was first assembled on a solderless breadboard. After confirming that everything was working correctly, the circuit was assembled on a perforated prototyping board, using a soldering iron and a 20 gauge wire for jumper connections. The lever was a 5 cm long × 0.5 cm wide × 0.1 cm thick piece of aluminum that was cut from a piece of aluminum with tin snips, hammered flat, and 1 mm diameter holes were drilled about 3 mm from each end.
Petal breakstrength assay
A standard curve (voltage as a function of weight) was generated by sequentially measuring the voltage generated with several objects (paper clips) of known weight suspended from the petal gripper. The voltage measured without any extra weights attached was subtracted from voltages of all weights, to correct for the weight contribution of the petal gripper and lever. The data were fitted to a linear curve using Microsoft Excel to obtain the y = mx + b equation. When an unknown (i.e. a petal pull) is measured, the observed voltage (after performing the aforementioned voltage subtraction from the weight of the petal gripper and lever alone) was used to calculate the gram equivalents of force using the standard curve. The petal pulling was performed manually. The maximum voltage value from the acquisition during petal pulling was used to calculate the petal breakstrength. We used 3.5 week old Arabidopsis thaliana Columbia ecotype plants that were grown at a temperature of 22°C temp with 16 h 100 μmolm-2sec-1 light/8 h dark. The petal could be seen in the jaws of the petal gripper following pulling, as a visual confirmation that the petal did not slip out or tear during the assay (Fig. 1E).
We thank Jason Doke for fabricating the lever and for ordering supplies. We thank the members of the Walker lab for comments on the manuscript. Support for SKC was provided by the Post-doctoral Fellowship Program of the Basic Research Promotion Fund from the Korean Research Foundation (M01-2004-000-10035-0). Funding to JCW was provided by NSF 2010.
- Patterson SE: Cutting loose. Abscission and dehiscence in Arabidopsis. Plant Physiol. 2001, 126: 494-500. 10.1104/pp.126.2.494PubMed CentralView ArticlePubMedGoogle Scholar
- Roberts JA, Elliot KA, Gonzalez-Carranza ZH: Abscission, dehiscence, and other cell separation processes. Annu Rev Plant Biol. 2002, 53: 131-58. 10.1146/annurev.arplant.53.092701.180236View ArticlePubMedGoogle Scholar
- Patterson SE, Bleecker AB: Ethylene-dependent and -independent processes associated with floral organ abscission in Arabidopsis. Plant Physiol. 2004, 134: 194-203. 10.1104/pp.103.028027PubMed CentralView ArticlePubMedGoogle Scholar
- Kandasamy MK, McKinney EC, Deal RB, Meagher RB: Arabidopsis ARP7 is an essential actin-related protein required for normal embryogenesis, plant architecture, and floral organ abscission. Plant Physiol. 2005, 138: 2019-32. 10.1104/pp.105.065326PubMed CentralView ArticlePubMedGoogle Scholar
- Jinn TL, Stone JM, Walker JC: HAESA, an Arabidopsis leucine-rich repeat receptor kinase, controls floral organ abscission. Genes Dev. 2000, 14: 108-117.PubMed CentralPubMedGoogle Scholar
- Butenko MA, Patterson SE, Grini PE, Stenvik G, Amundsen SS, Mandal A, Aalen RB: INFLORESCENCE DEFICIENT IN ABSCISSION Controls Floral Organ Abscission in Arabidopsis and Identifies a Novel Family of Putative Ligands in Plants. Plant Cell. 2003, 15: 2296-2307. 10.1105/tpc.014365PubMed CentralView ArticlePubMedGoogle Scholar
- Hepworth SR, Zhang Y, McKim S, Li X, Haughn GW: BLADE-ON-PETIOLE-dependent signaling controls leaf and floral organ patterning in Arabidopsis. Plant Cell. 2005, 17: 1434-4. 10.1105/tpc.104.030536PubMed CentralView ArticlePubMedGoogle Scholar
- Ellis CM, Nagpal P, Young JC, Hagen G, Guilfoyle TJ, Reed JW: AUXIN RESPONSE FACTOR1 and AUXIN RESPONSE FACTOR2 regulate senescence and floral organ abscission in Arabidopsis thaliana. Development. 2005, 132: 4563-74. 10.1242/dev.02012View ArticlePubMedGoogle Scholar
- Okushima Y, Mitina I, Quach HL, Theologis A: AUXIN RESPONSE FACTOR 2 (ARF2): a pleiotropic developmental regulator. Plant J. 2005, 43: 29-46. 10.1111/j.1365-313X.2005.02426.xView ArticlePubMedGoogle Scholar
- Fernandez DE, Heck GR, Perry SE, Patterson SE, Bleecker AB, Fang SC: The embryo MADS domain factor AGL15 acts postembryonically: Inhibition of perianth senescence and abscission via constitutive expression. Plant Cell. 2000, 12: 183-98. 10.1105/tpc.12.2.183PubMed CentralView ArticlePubMedGoogle Scholar
- Patterson SE: Characterization of delayed floral organ abscission and cell separation in Arabidopsis thaliana L. Heynh PhD thesis. 1998, University of Wisconsin, Madison.Google Scholar
- Smyth DR, Bowman JL, Meyerowitz EM: Early flower development in Arabidopsis. Plant Cell. 1990, 2: 755-67. 10.1105/tpc.2.8.755PubMed CentralView ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.